The following photos were taken using a custom pulse burst generator that allows a high frequency burst of 1 nanosecond pulses at differing repetition retes to be put into the PV16 to verify function.
The top white trace is the output of he test pulse generator as seen by a 3 GHz Tektronix oscilloscope.
The second green trace shows the gate width.
The bottom blue trace is the output of the PV16
The trace overlap is determined by the length of the cables, and there is some small distortion presented to the PV16 because the test pulser is split into the oscilloscope and the PV16
900 MHz burst of 10 ns width at 367 Hz
900 MHz burst of 64 ns width at 367 Hz
900 MHz burst of 488 ns width at 367 Hz
900 MHz burst of 10.98 ns width at 6.77 MHz repetition
900 MHz burst of 10.47 ns width at 367 Hz
100 MHz burst of 9.9 ns width at 367 Hz
200 MHz burst of 9.91ns width at 367 Hz
500 MHz burst of 10 ns width at 367 Hz
500 MHz burst of 67 ns width at 6.73 MHz repetition rate